|Place of Origin:||Dongguan China (Mainland)|
|Minimum Order Quantity:||5 UNITS|
|Packaging Details:||Aluminum Case, Sterilization Case|
|Delivery Time:||Within one week after receiving 100% prepayment|
|Payment Terms:||T/T, Western Union|
|Supply Ability:||200-300 UNITS / MONTH|
|Properties:||Orthopedic Power Tools||Type:||Surgical Power Tools|
|Product Name:||Surgical Power Normal Drill||Battery:||Ni-MH Battery|
|Voltage:||14.4V||Battery Charged:||110V / 220V|
|Materail:||Aluminum||Package:||Aluminum Case, Sterilization Case|
|Battery Capacity:||1600mAh||Shipping Time:||5 Days|
medical power tools,
ortho drill medical equipment
Applicable To Provide Mechanical Power For Bone Tissue Surgery,Electric Drill For Medical Electric Tools
1. Max Rotational Speed: 1200rpm
2. Green Batteries: 14.4V
3. Stepless, Offer Reversing Choice
4. To Clamp Drill Bit for φ 0.6-8mm
5. Host Powerful, Siamese Closed Back Cover
6. Positive and Negative Ratation, smooth operation and stepless speed regulation
7. Environmental protection battery within voltage of 7.2 . No need for sterilization8.
9. Whole Machine Can Be Autoclavable And Sterilized By High Temperature
10. Battery charged suitable for voltage of both 110V and 220V
11. Complete Set: One motor , one chuck , two batteries , one charger , two separate sterilize channel
Packaging & Delivery
Packaging Details:White color box, sterilization case, aluminum case
Lead Time : Within one week after receiving 100% prepayment
Medical electric surgical drills are mainly used in drilling and steel plate screws in trauma surgery. Made of high-strength biomedical aluminum, its perfect ergonomic design, infinitely variable speed, low noise, low vibration and easy to use give the operator and patient a good experience. 4.5mm hollow, suitable for various bone fixing operation easily and conveniently. can be high temperature and high-pressure sterilization, safe and reliable performance reduces the risk of surgery.
|5||Two Separate Sterilize Channel|
|6||One Operating Manual|
|7||One Guarantee Card|
Instructions for use
1. Electric bone drill can not be disassembled or altered at will, and can not be tapped at will to avoid affecting its performance.
2. Make sure the bit is firmly stuck before drilling.
1. Advanced motor technology.
2. Main components are made of Aluminum
3. Reduced vibration and enhanced handpiece control.
4. Long working time for battery
Matters needing attention
1. It is strictly forbidden to overuse surgical instruments.
2. It is strictly forbidden to use corrosive solvents to wipe and immerse the instruments and sterilize them.
3. In the process of using the electric bone drill, we should prevent the water from entering the battery box, damaging the battery or causing short circuit of the electrical appliances.
1. When the battery can not be recharged, it must be sent to the Equipment Division for repair immediately.
2. If you find that the speed of the electric drill can not be controlled, you must immediately notify the relevant maintenance personnel of the equipment department to attend the test.
3. If the drill bit is worn, it must be replaced so as not to affect the quality of the operation.
6. Regularly check battery charging and battery adequacy.
1. Our whole power tool can be autoclavable for 135 °C with 1.6Mpa, in this way the bacilli and virus which can not killed by the suffocating will be killed
2. We use environmental friendly nickel-metal hydride battery
3. The machine has very low noise, High precision
4. Our products has one year warranty period
5. Ergonomic Design, Portable, Lightweight, Durable
1. We are manufacturer and have our own factory
2. Customized design is avaiable, OEM & ODM are welcomed
3. Best after service: 1year warranty and lifetime service
4. Prefession and Hardworking Team, reply your inquiry in 24 hours
5. Quality assurance, Traumedtech is a strong enterprise
The handpiece guarantees for one year, the battery and other fast consumption items guarantee for 3 months.
1.Sterilize temperature ≤ 135 °C, if the temperature is over 135 °C, the motor will be damaged.
2.The battery need to be taken out from the power tool after surgery. Otherwise the small automatic discharge of the electric will make the life of the battery short.
3.The battery cannot be sterilized, it is not autoclavable.
4.The battery space in the handpiece of the tool should not have any water,after sterilization.
5.The dill bit must be washed right after the surgery by purger in order to aviod the blood stick in the dill bit, then we should make it dry.
Q: Trading company or manufacturer?
A: We're a manufacturing company. Our location is in Dongguan, Guangdong Province, China. 0.5 hours by high-speed train from Shengzhen.
Q: Regions of sales?
A: At present our main overseas markets are Asia, South America and Europe.
Q: Regional Distribution?
A: Yes we are looking for and welcome distribution.
Please contact us for discussion if you are interested to become our distributor.
Q: How long is delivery time?
A: Generally 7-15 days if stock available and 30-45 days if regular production needed.
Depending on order quantity, product type, and other order factors and requirements.
Q: Samples available? Free or extra ?
A: We can accept sample order. With extra charge.
Q: Terms of payment ?
A: T/T,Western Union.
Q: Any OEM service?
A: We can offer OEM service. Products can be made-to-order and with special requirements.
Full name: Traumed Technology Co., Ltd
Address: 5th Floor, Building C, Reyuan Industrial Park, No.28 Qingtang Road, Youganpu Village Fenggang Town, Dongguan City, Guangdong Province, China
Post Code :102605
Website : www.traumedtech.com/en
Contact Person: Mr. Stanley Liu